I've been trying to complete the FT calibration using a new Fluke QA ES III, whilst carrying out the "Current Calibration" at 5A using the vessel sealing mode on the ES III , the Fluke starts to frequently time out, reading only
In Debug mode the FT responds as it should.
Has anyone had similar problems or should I think about replacing the RF board ??
Still waiting to try another FT which the dept. will not release ....